Acu-Rite S Manual Machine Tool Digital Readouts for Milling and Turning. Acu-Rite S DRO packages with Linear Encoders for almost any manual. Find great deals on eBay for ACU Rite DRO in Digital Readouts for DRO; DROM, DROE; VRO; S; S; Acurite VUE; Anilam Wizard ;. Acu-Rite S DRO. The ACU-RITE series with LCD screen technology offers the possibility of running different types of operations with just one readout.
|Published (Last):||1 November 2008|
|PDF File Size:||10.94 Mb|
|ePub File Size:||19.88 Mb|
|Price:||Free* [*Free Regsitration Required]|
The Measured Field shows 0. Working without reference mark The S can also be used without crossing over the reference marks. Scale Factor Scale Factor The scale factor may be used to scale the part up or down. Thanks for the feedback guys and Jon thanks for the tip about the new DRO’s. Just guess and then check for accuracy and adjust if the DRO doesn’t display accurate measurements?
The Tool Description form will appear. The graphic in the message box aids in selecting the correct plane. You can calculate tapers either by entering dimensions from a print, or by touching a tapered workpiece with a tool or indicator.
Browse Related Browse Related. To print the manual completely, please, download it. I detest membrane keypads.
Page 74 II – 1 Installation Setup Linear Error Compensation Linear error compensation can be applied, if the results of the comparison with a reference standard show a linear deviation over the whole measuring length.
Probing function Circle Center First center point coordinate, e. Touch the workpiece face with the tool. Refer to screen examples on the right. To calculate the linear error compensation use this formula: These forms will appear after selecting features that require any additional information. Serial Port A printer or computer may be connected to the serial port. I – 1 Fundamentals of Positioning Datums The workpiece drawing identifies a certain point on the workpiece example: Also output of the current display positions is activated via a command Ctrl Bsent acu–rite the 200s over the serial port.
Enter data and execute a circle pattern. Both values are in the units indicated in the form. The acu-ritte of the centerline M is determined by probing the acy-rite 1, and 2. The preset Distance-To-Go value shown is the previous preset value entered.
I – 4 Turning Zcu-rite Operations Presets The functionality has been explained previously in this manual see “Presets” on page The following soft keys are available while in the Milling Feature table. More expensive of course This may not reflect the user defined count direction, and is only required for determining the non-linear error compensation.
Data which can be exported from the S to an external serial device includes: Page 22 I – 2 Auc-rite Operations for S Help Screen The integrated operating instructions provide information and assistance in any situation. I – 1 Fundamentals of Positioning I – 1 Fundamentals of Positioning Datums The workpiece drawing identifies a certain point on the workpiece example: I – 4 Acu-ritr Specific Operations Vectoring Vectoring breaks down the movement of the compound axis into the crossfeed or longitudinal axes.
Retract the edge finder from the workpiece. See screen to the right. Used to divide the current position by two page I haven’t been able to find any identification numbers on the two scales on my mill.
I – 3 Milling Specific Operations Probing with a Tool If using a tool, or non-electrical edge finder to set datum points, the S probing functions can still be used. Page 13 Features Milling Acu-rite number Male AR 9 pin connector to female round 6 pin connector.
200S Readout System
It is defined as one of the two axes in the plane of rotation. Page 78 II – 1 Installation Setup To measure the standard using an edge finder, touch one end of the block. Table of Contents Add to my manuals Add.
Desired coordinate of the centerline: The current selection is shown on the soft key, and in the plane field. Once the new nominal position information is entered the display will switch to Distance-To-Go mode and show the distance between the current position and the nominal position. The explanation, and examples on those pages are based on a mill application.
Again, they advertise it, but I have never seen it done myself.
ACU-RITE S : User guide
Acu-rit Reference Marks Encoders normally contain one or more reference marks which the S Reference Mark Evaluation feature uses to re-establish datum positions after a power interruption. Once the new nominal position information is entered, the display will switch to Distance-To-Go mode. Press the Down arrow key until the X axis 20s is highlighted. Press C to exit. Skip to main content. Afu-rite soft key, and enter the passcode. They havent been released yet, but they look sweet as all get out.
So this leads me to believe that both scales are operating normally and the issue is with the DRO unit itself. Not sure if they rubbed off over time? II – 1 Installation Setup To measure the standard using an edge finder, touch one end of the block.