Question regarding a FLuke A meter display. «on: January 28, , 04 am». So I’ve gotten into restoring “vintage” DMMs from Ebay. I’ve gotten. Download FLUKE A IM.V4 SERVICE MANUAL service manual & repair info for electronics experts. Hi guys. I managed to buy a Fluke A for £1, knowing that it wasn’t working. The selection switches were broken bar one, and they were.
|Published (Last):||2 March 2007|
|PDF File Size:||14.46 Mb|
|ePub File Size:||20.45 Mb|
|Price:||Free* [*Free Regsitration Required]|
Autoranging Test Installation instructions are supplied with the kit, and appear in Section 6 of this manual. Noting the range or ranges affected by the malfunction may point to the gain control component or flukr circuitry causing the problem.
Insert the open end of the case into the rear opening on the center rack mount plate. See details for description of any imperfections. H K Ltd 28th FI. Control and Display Select the AUTO pushbutton.
Fluke A for Sale|Voltmeters|Meters|Test Equipment Center
Parts lists include the following information: All dc voltage ranges will withstand the continuous applica- tion of inputs up to V dc or 1 V peak ac. Repeat steps 1 8 and 1 9 of this procedure until the dc voltmeter reading is between.
To ensure prompt and efficient handling of your order, include the following information. Senen Raya 44, P. Cleveland, Ohio The Budd Co. Remove the A from the outer case. If the compare pulse is not present or in- correct, either in amplitude or timing of the leading edge make the following check.
The A display should read within the limits shown. Carlisle, Pennsylvania Electro-Midland Corp.
The listing of this manual is to help those who need this manual to repair and maintain their equipment. Each instrument manufactured by the John Fluke Mfg. Box Wellington, New Zealand Tel: This provides access to the calibration adjust- ments. The A D Converter operation can be checked using the following procedure. Complete the calibration adjustments procedure if the AC Converter fails any part of this test.
Carefully inspect the interior of the instru- ment: Select the V range.
Repairing a Fluke 8810A Multimeter
The recommended test equipment for troubleshooting is listed in Table Once installed in the A, the DOU re- quires no operator attention. All pcb assemblies are documented at their fpuke revision level.
Full autoranging or manual ranging Overload Protection. New Hampshire Johanson Mfg. Resistor Selection Procedure A low reference supply would also cause period B to lengthen. If the recom- mended equipment is not available, equipment of equivalent specifications may be used.
Fluke 8810A Digital Multimeter Missing Rubber Foot Pad – Used
Because the timing of the control signals used to process the input signal through the A D Conver- ter and display circuits is critical to the proper operation of these sections, an oscilloscope should be used to make the voltage and signal checks. Selecting the volt range in the Fkuke function, in addition to configuring the differential amplifier circuit for unity gain, closes relay K5 to provide a The performance test is designed to compare the A performance with flukr accuracy specifications given in Section 1 of this manual.
The ground lead on the power cord is connected to chassis ground through the A power supply and should be connected to a high quality earth ground.