IEC provides procedures for calibrating multimode optical time domain reflectometers (OTDR). It covers OTDR measurement. Buy IEC CALIBRATION OF OPTICAL TIME-DOMAIN REFLECTOMETERS (OTDR) – PART 2: OTDR FOR MULTIMODE FIBRES from SAI Global. Buy IEC CALIBRATION OF OPTICAL TIME-DOMAIN REFLECTOMETERS (OTDR) – PART 1: OTDR FOR SINGLE MODE FIBRES from SAI Global.
|Published (Last):||14 September 2012|
|PDF File Size:||9.21 Mb|
|ePub File Size:||3.73 Mb|
|Price:||Free* [*Free Regsitration Required]|
Although we endeavour to maintain an up-to-date catalogue, on occasion we may not have the latest version of a document, including where it has not yet been provided through an international feed.
This standard is not included in any packages.
If you need help with the purchase or have any queries please email enquiries standards. A ghost will not have any loss, it will be at equal distance from a highly reflective event look for repetitiontends to 617746 in the middle of noise after the end of the cable. Are there some reusable connectors in the commerce?
IEC – European Standards
If you are using an OTDR without training, you are going to have big problems. Add to Alert PDF. A knowledgeable contractor should be able to educate the user on proper testing requirements.
It is eic considered a replacement for insertion loss testing by knowledgeable fiber optic personnel. Best to iev sure the trace is not noisy to the end and have dB from the cable backscatter level to the noise floor. I am testing a cable with OTDR.
It constitutes a technical revision. First, what are the sources of error? OTDRs are always used with a launch cable and may use a receive cable.
Results are archived 617746 other documentation to be available if restoration is necessary in the future. OTDR for single mode fibres This document has been re-assessed by the committee, and judged to still be up to date.
Splice threshold Reflectance threshold Slope threshold slope is attenuation coefficient End threshold depends on whether you 1 use receive reference cable which would be a normal connection loss or 2 the length of the cable and the noise floor of the measurement.
Your Alert Profile lists the documents that will be ief.
It only covers OTDR iiec errors and uncertainties. Some multimode OTDRs are now usable for short length multimode premises cables but only if they are properly set up before use and launch and receive cables have connections with low reflectance. If you shoot in both directions and overlay, the software should recognize that there should be events in both directions, input a “0 dB” event and average accordingly.
IEC 61746-1 Ed. 1.0 en(2009)
International Classification for Standards ICS is an international classification system for technical standards. The launch cable, sometimes also called a “pulse suppressor,” has two major reasons for its use: Premises cabling, 617446, has short cable runs and almost never includes splices, so the requirement of OTDR testing appears to be as an alternative to insertion loss testing with a light source and power meter, which in reality, is inappropriate.
How can we differentiate a ghost from a real event? For this reason alone, we recommend that insertion loss testing be done even when OTDR testing is required by installation contracts.
There are many times a splice is undetectable in an OTDR trace due to good splices and the simple fact that the OTDR measurement technique itself is limited. The actual resolution of the OTDR is usually thought of as its ability to distinguish between two points on the cable, like intermediate patchcords or closely spaced splices.
Oec OTDR uses the speed of light in the fiber from the index of refraction to calculate the length of the fiber. Follow these links to find other Standards similar to the one you are viewing.
GSO IEC – Standards Store – GCC Standardization Organization
An insertion loss test made with a light source and power meter is a simple test that is similar in principle to how a fiber optic link works. I changed the pulse width but nothing happened. I have a limit 0. See the demonstration below for a way to prove this to yourself. Most of the technical calls we get at FOA regarding problems using OTDRs on premises cabling systems are caused by users who don’t know what an OTDR is requiring them for testing their installation just because somebody told them to or they assume a bigger, more expensive instrument gives better data.
For a splice with 0. Have a long pigtail on the OTDR as a launch cable, long enough for the test pulse to settle, say m, then use a splice for a temporary connection. Standards Subsctiption may be the perfect solution. Also a short connection cable attached to the OTDR before the launch cable that never gets removed from the OTDR prevents excess wear on the panel connector.
You may delete a document from your Alert Profile at any time. The large event you see right in front of the instrument on the OTDR trace is caused by crosstalk within the instrument and reflectance from the connector on the face of the OTDR.
Directional Results Can Be Confusing: Please first verify your email before subscribing to alerts. New international testing standards, however, accept both OTDR testing and insertion loss testing, even for short premises multimode cable plants.
A receive cable may be used on the far end to allow measurements of the connector on the end of the cable under test also. I use bi directional analysis. The answer to this question has several aspects.